The NanoTube N2 X-ray source consists of the head with dimensions 646 x 220 x 372 mm (L x W x H) excluding cables. The head has several mounting holes consistent with optic table mounting.
In addition, the NanoTube N2 consists of two 4U (176 mm height) 19” rack electronics boxes, floor standing generator and a 19” rack mounted chiller, that can be mounted up to 5m from the head.
Source head dimensions
Installation and operation
The NanoTube N2 can be remotely operated through TCP/IP or directly through the GUI. The GUI can be operated on the source itself if it is equipped with monitor, keyboard and mouse, or on most computer platforms with a TCP/ IP connection to the source.
The source cannot be operated as a standalone unit and must be integrated into a system providing the proper interlock connections.
Mains: AC, single phase, 200-240 V, 16 A, 50/60 Hz. Ambient: 20-25°C (stable within ± 0.2°C for optimal source stability), max 85% relative humidity.
The cookie settings on this website are set to "allow cookies" to give you the best browsing experience possible. If you continue to use this website without changing your cookie settings or you click "Accept", you are consenting to this.